|Using the Elcometer 134A Chloride Ion Test Kit for Abrasives in the field will accurately identify contamination and prevent costly surface-related failures. Chlorides deposited on a surface by contaminated abrasives during blasting can cause a coating to fail prematurely. Contamination can build up, particularly if the blast media is recycled several times.|
Elcometer 134A Chloride Ion Test Kit for Abrasives – Specifications
|E134—-2||Elcometer 134A Chloride Ion Test Kit for Abrasives|
|Measurement Range||1 – 60µg/cm² (1 – 60ppm)|
|Scale Resolution||1µg/cm² (1ppm)|
|Sample Time||1.5 minutes (approx)|
|Tests per kit||4|
|Storage Conditions||Not exceeding 25°C|
|Dimensions||185 x 125 x 110mm|
Elcometer 134A Chloride Ion Test Kit for Abrasives Packing List:
- 4 x test kits
- containing: abrasive sample container
- mixing container with a pre-measured quantity of solution
- titration tube
- titration tube snapper
- operating instructions
Elcometer 134A Chloride Ion Test Kit for Abrasives – Data Sheet
Elcometer 134A Chloride Ion Test Kit for Abrasives – Test Method
The Elcometer 134A is simple and convenient to use with its three-step process
1. Pour CHLOR*EXTRACT™ solution into the latex tube.
2. Peel the protective backing off the flange of the tube, pinch to tube to retain the CHLOR*EXTRACT™ and stick the flange to the surface to be tested and work the solution against the surface to extract the salts.
Peel the flange off. Insert the glass titration tube into the solution in the tube and read the result.